Complex, intraarticular, proximal ulnar fracture dislocation, with associated distal ulnar shaft fracture.
CT showing complex fracture dislocation of proximal ulna
3D CT showing complex fracture dislocation.
A direct posterior approach is used, allowing exposure from the olecranon to the distal ulna because of the segmental fracture.
The complex proximal fracture is exposed, and the fragments identified.
This image shows the cartilage damage after such a severe elbow injury, and one piece of unattached osteochondral bone being removed.
This direct posterior approach allows for the easy visualization of the coronoid fracture which in this instance is very comminuted. Here a K-wire is being placed for reconstruction of the comminuted coronoid.
Here a K-wire is being used to reconstruct the surface of the coronoid before definitive bridging fixation.
Once the coronoid is reconstructed the olecranon can be reduced to the shaft.
The olecranon is temporarily held with K-wires.
This lateral X-ray shows a single lag screw into the coronoid along with multiple K-wires temporarily holding the fracture dislocation reduced.
The proximal ulnar LCP is applied and the plate is firmly held against the ulna to help in the reduction of the comminuted proximal ulna.
A variable angle screw is inserted into the proximal ulna to supplement the K-wire fixation.
This plate does not have many points for proximal fixation, so supplementary K-wire fixation is advised as well.
Once the ulna is brought out to length, the fracture is bridged and a screw is applied to maintain length.
This image shows the proximal ulna reduced and bridged with both proximal and distal fixation.
AP image showing the coronoid reduction.
Lateral image showing satisfactory reconstruction of the proximal ulna, but the radial head is not completely reduced in the radiocapitellar joint.
A distal incision is made to expose the distal ulnar shaft fracture. If necessary, this incision can be extended proximally to join with the distal end of the previous incision.
The fracture is held reduced with a clamp and a hole is drilled for lag screw.
The hole is countersunk and a lag screw inserted.
A large reconstruction plate was selected and applied in neutralization mode.
Screws are inserted in neutral mode.
Note there is only a small gap between the two plates, which creates a stress riser. This could have been avoided if a shorter proximal VA-LCP plate had been selected. Alternatively, a longer distal plate could have been chosen so that the two plates overlapped.
Final image prior to closure
Lateral X-ray showing fixation of both fractures. The radial head is now reduced.
AP of elbow and forearm showing adequate reduction of the proximal ulna
7 week clinical images showing gradual return of elbow function
1.5 years AP and lateral X-rays showing complete healing
Final, 1.5 year clinical images showing slight loss of extension, but otherwise good elbow function.