1 General considerations top
Reamed versus unreamed nailing
In general, reamed nailing allows the use of larger diameter implants and may therefore provide greater initial stability. Some studies show a tendency towards less malunion and nonunion when reaming is performed. On the other hand, fat embolization is a concern and with the solid unreamed nails, the forces required to achieve nail torsion are higher. Most centers prefer reamed nailing as the standard procedure.
Important module-wide statement
Where appropriate, a “generic” fracture zone will be illustrated and not necessarily the specific fracture morphology under consideration. Where the fracture morphology determines the fixation technique, the specific morphology will be shown.
2 Reduction topenlarge
In more proximal fractures, due to the pull of the iliopsoas muscle, the upper main fragment may be flexed and externally rotated, and the distal segment lies posteriorly due to gravity.
Several options for fracture reduction can be considered:
- Elevation of the distal fragment by use of a crutch.
- Lowering of the proximal fragment by external pressure from a mallet.
- A wrap around the femur.
- A Schanz screw inserted into one of the fragments.
- Use of a bone hook.
- (Manual reduction)
The proximal fragment may be flexed and externally rotated by the iliopsoas muscle.
Elevation of the distal fragment by use of a crutch
This option may only be used if the patient is on traction. A crutch is slid beneath the distal main fragment in order to elevate it to the level of the proximal fragment.
Lowering of the proximal fragment by external pressure from a mallet
Firm manual pressure is usually required to achieve fracture reduction.
A wrap around the femur
Based on the nature of the fracture, the wrap is usually placed around the larger fragment.
Reducing the fracture using a wrap.
A Schanz screw inserted into one of the fragments
A monocortical Schanz screw (preferably 5 mm) can be helpful for providing direct control of the displaced fragments. It is superior to reduction maneuvers through the skin.
Direct reduction with a bone hook may be helpful for securing an anatomic alignment. Careful insertion and manipulation must be performed, in order to minimize soft-tissue trauma and to prevent injury to the femoral artery.
Reduction using a bone hook.
Manual reduction may be attempted, but radiation of the surgeon‘s hands may be unavoidable.
Guide wire insertion
A guide wire is advanced into the distal main fragment until it is about 5 mm proximal to the intercondylar notch. It is important that the guide wire be centered in order to prevent eccentric reaming and subsequent malposition of the nail, which can result in varus/valgus/antecurvatum/retrocurvatum malalignment.
Unreamed nailing does not require a guide wire. When unreamed nailing is performed, the nail is used as a reduction tool.
To ensure maintenance of alignment of the K-wire throughout the reaming process, it may be gently be tapped in order to provide purchase in the cancellous subchondral bone. If this is not achieved, the guide wire may displace on removal and exchange of the reamers.
3 Poller screw (blocking screw) topenlarge
The concept of a poller screw is based on the principle that the malalignment induced by oblique, proximal and distal fractures can be counteracted by the nail-directing effect of the screw. Therefore, its position should aim to counteract the displacement of the fracture. Most often it is therefore placed in the short side in the distal fragment. If there is a wide canal, two screws can be placed on either side of the path of the nail.
It is advisable to insert the poller screw prior to reaming, and in case of unreamed nailing, prior to nail insertion, in order to provide adequate contact between the nail and the screw. If the poller screw is inserted after reaming, the path of the nail is already set and the poller screw may not work. A small fracture screw or a locking screw can be used as a poller screw, depending on the local anatomy.
The reaming process in the presence of the poller screw must be performed very carefully in order not to harm the reamer tip.
The most frequent indication for poller screws is a proximal or distal oblique fracture which tends to shift when the axial knee blow technique is used to close any fracture gap, or when the patient is mobilized.
In these cases, prophylactic poller screws are very helpful.
Stedtfeld HW, Mittlmeier T, Landgraf P et al (2004) The Logic and Clinical Applications of Blocking Screws. J. Bone Joint Surg. Am. 86:17-25, 2004.
Midshaft comminuted fractures
Poller screws are used according to the initial displacement and the size of the canal. In general, midshaft fractures tend to align better than more proximal or distal fractures and, therefore, the indication for poller screws is uncommon.
Proximal oblique fractures
Proximal fractures tend to fall into a varus malalignment. Therefore, in most cases a medial blocking screw helps to counteract this problem. When applying this technique, care should be taken not to cause any damage to the neurovascular structures.
Distal oblique fractures
The muscular forces in oblique fractures try to displace the fracture. The blocking screw counteracts these forces when properly positioned on the short side of the distal fragment.
4 Determine nail length and diameter topenlarge
Determine nail length via guide wire (only with reamed nailing)
The correct length of the nail is determined by comparing a second guide wire to the one that has been inserted. The correct placement of the guide wire in the distal canal should be assessed via image intensifier. Additionally, the second guide wire must be positioned in contact with the greater trochanter. This must be verified by image intensifier as well.
Alternatively, a radiographic ruler may be used.
The tip of the ruler should be positioned in the center of the distal end of the femur. Nail length is determined by the position of piriformis fossa, not by the tip of the greater trochanter.
It is important to visualize the fracture zone by image intensifier to ensure that adequate restoration of femoral length has been achieved.
It is important to measure the medullary diameter at the mid portion of the femur, which represents the narrowest segment of the medullary canal.
The inner cortical edge should touch with the inner numbered disk of the ruler aperture. In the illustration an inner cortical diameter of 14 mm is shown.
Consideration for special situations
In multifragmentary fractures, or in open fractures with bone loss, it is safer to perform preoperative planning on the uninjured leg. In bilateral femoral fractures, the less comminuted side should be used to determine the length and diameter of the nail.
5 Reaming topenlarge
Insertion of reaming rod
After the tissue protector has been introduced, the reaming shaft, fitted with the first reamer head, is inserted over the guide wire. Usually reaming begins with a 9 mm medullary reamer.
Sequential reamer size increase
Reaming is performed in sequential steps by increments of 0.5 mm each.
As soon as chatter from cancellous bone can be felt and heard, the inner cortex has been reached. This may not be the case in segmental fractures or when severe comminution is present.
Adequate reaming must be performed in order to allow for smooth nail insertion. For example, for a nail width of 10 mm, drill bits of up to 10.5 or 11 mm diameter are used. If a very tight fit of the reamer can be detected before the desired reaming size is reached, one should consider using a smaller nail than previously planned.
Pitfalls: eccentric and overaggressive reaming
Eccentric reaming can cause weakening of the adjacent cortex which may interfere with healing or even cause a fatigue fracture.
Trapping of reamer by slow spinning
If the reamer becomes trapped while reaming, it must be gently removed by the most senior surgeon, because breakage of the reamer tip in this situation can be a devastating complication.
Heat necrosis by overaggressive reaming
Overaggressive reaming should be avoided because it may cause heat necrosis of the femoral canal. This applies especially for narrow midshaft canals (9 mm or less in diameter).
Rapid thrusting/systemic fat embolization
Care should be taken to use sharp reamers, to advance the reamers slowly, and to allow sufficient time between reaming steps in order for the intramedullary pressure to normalize. Rapid thrusting of the reamer may worsen the intramedullary pressure increase that is observed during nailing. This image demonstrates fat extrusion in a human cadaver specimen with a window in the proximal section.
This may cause pulmonary embolization of medullary fat, which in turn may lead to pulmonary dysfunction (lower image in the enlarged view shows an example of fat embolization through the right atrium).
Special situation: conversion from an external fixator to an intramedullary nail
Because the external fixator is still in place, the Schanz screws must be partially withdrawn to allow the guide wire, the reamers, and later the nail, to pass through. The external fixator is held in place by monocortical purchase to assure that the fracture remains stable. The external fixator also acts as a joystick for the reduction.
6 Nail insertion topenlarge
Connecting handle to nail
The insertion handle is connected to the nail by the corresponding connecting screw. It is attached using the hexagonal screwdriver through the hole in the insertion handle. It is recommended that the nail be inserted manually and rotated about 90 degrees from its point of entry to its final orientation.
Introduction of nail
Under control with the image intensifier, the nail is pushed down as far as the fracture zone. After the driving cap has been fixed to the insertion handle, the nail is further advanced into the medullary cavity by gentle hammer blows, whilst verifying the position of the tip of the nail under the image intensifier.
In this intraoperative view, the nail is about to pass the fracture site in the intensifier image.
In this intraoperative view an unreamed nail is inserted and the external fixator is used as a joystick to help to reduce the fracture anatomically.
Once past the fracture, the nail is advanced by hand or by gentle hammer blows.
Passing the fracture zone
It is important that the tip of the nail does not become trapped in the distal main fragment because blow out fractures can occur. Each gentle hammer blow should advance the nail. Do not force the nail through a tight canal – if necessary, re-ream to another 0.5 mm diameter.
In many cases, the nail will help to align the fracture.
A solid nail (unreamed technique) passes the fracture site into the distal medullary canal.
Assessing proper nail insertion
The nail is then completely inserted. This is assessed by using an additional K-wire that marks the upper end of the nail.
7 Nail locking topenlarge
Purpose of locking
Locking was developed in order to provide and maintain rotational stability and length. It can also be used to finalize fracture reduction.
Type of locking
For transverse fractures, dynamic locking may be performed, otherwise static locking is advisable as shown in the illustration.
The options available with regard to the proximal locking will depend on the implant chosen.
C-type fractures are the least axially stable fractures. Special care must be taken to avoid limb shortening. The length may be assessed by comparison to the contralateral femur.
8 Distal locking topenlarge
Verification of nail position
Before locking, the correct position of the nail and the rotation of the femur must be verified.
If no traction table is used (ie, using the freehand technique) the cable method may be used. In this approach, a line is drawn from the iliac spine across the patella to the cleft between the first and second toes. If rotation is correct, this line will pass over the mid line of the patella. The radiological landmarks of the center of the femoral head, the center of the knee and the center of the ankle joint should all be in line if the mechanical axis of the femur is correct.
Another method of assessing rotational reduction is to compare the cortical thickness above and below the fracture. If a shaft fracture is multifragmentary, the image intensifier cannot be used to assess the analogue cortical diameter on both sides of the fracture.
For distal locking, the image intensifier is brought into a strict lateral position. The distal hole must be seen as a perfect circle and the tip of a scalpel is projected into the center of the hole.
This guides the stab incision down to the femoral cortex.
Screw insertion technique
The radiolucent drive helps to position the drill bit so that the locking screw can be properly inserted. While the locking hole is drilled, the assisting surgeon must prevent the lower extremity from moving in order not to miss the target hole. The length of the locking screw is determined, using the appropriate depth gauge. Then the locking screw is inserted.
If a radiolucent drive is not available, the projection of the tip of the drill bit should be placed as centrally as possible (see image). Start drilling but assess the position of the tip of the drill bit repeatedly, with the drill temporarily uncoupled.
Pearl: secure screw using a suture
If the contact between the screw driver and the locking screw is lost, the
screw may move within the soft tissue and become extremely hard to capture. To
prevent this time-consuming complication, the locking screw should be lassoed
with a strong absorbable suture.
Pitfall: trapped locking screw
If the screw holes are not perpendicular to the nail, the locking screw may become trapped and may not be advanced properly.
Second locking screw
The second locking screw is inserted into the distal locking hole. After distal locking, an axial blow to the knee region may be be used to reduce any fracture distraction.
Alternatively, when distal locking is completed prior to proximal locking, a slotted hammer can be used to pull back the locked nail and the distal fragment.
Completed distal locking
The image shows the distal locking completed.
Pitfall: screw loosening
Care should be taken to capture the far cortex in order to prevent toggling when the patient is mobilized. This would lead to early loosening of the locking screw. In the image shown, the proximal locking screw is of correct length, but the distal one is marginally too short.
9 Intraoperative radiological assessment topenlarge
Assessment of rotation
The profile of the lesser trochanter is compared with that of the contralateral leg (lesser trochanter shape sign).
Before positioning the patient, the profile of the lesser trochanter of the intact opposite side (patella facing anterior) is stored in the image intensifier.
The illustration shows the lesser trochanter of the intact opposite side.
In cases of malrotation, the lesser trochanter is of different profile when compared to that of the contralateral leg.
Care should be taken to assess rotation with the patella facing directly forwards.
Matching of the lesser trochanter shape
After distal locking, the correction is achieved by using the handle of the nail insertion device. Thereby the distal main fragment can be rotated in relation to the proximal main fragment.
Cortical step sign and diameter difference sign
Cortical step sign
The presence of considerable rotational deformity may be diagnosed by the difference in thickness of the cortices above and below the fracture zone.
Diameter difference sign
The diameter difference sign is of use at levels where the bone cross section is oval rather than round. In cases of malrotation, the proximal and distal main fragments will appear to be of different diameters.
10 Proximal locking topenlarge
Proximal aiming device
If proximal locking is performed before distal locking, it is important to verify the correct position of the distal end of the nail under the image intensifier, before the proximal aiming device is attached to the insertion handle.
Preparation for screw insertion
The sites of the skin and fascial incisions for the drill bit and locking screws may be determined after the drill sleeve assembly has been in the holes of the aiming device.
The length of the locking screws is read from the calibrated drill bit. The correct length is confirmed antero-posteriorly using the image intensifier. The insertion handle should not be removed until the correct placement of all the locking screws has been verified.
Drill sleeve to bone contact
The drill sleeve must be in close contact with the bone at all times since it is important for the measurement of the locking screws. Close contact is maintained by pressing down on the sleeve with the contralateral thumb.
Verification of locking screw placement
A final x-ray in two planes should be obtained in all cases in order to verify the exact locking screw placement.
Special situation: associated femoral neck fracture
The "miss-a-nail" system has been designed for associated femoral neck fractures.
Special situation: segmental fracture with extension to the subtrochanteric region
In cases of segmental fractures with extension to the subtrochanteric region the standard femoral nail with a helical blade may be used.
Special situation: extension of fracture to the lesser trochanter
When a dedicated cephalomedullary device is not available and the fracture configuration permits, a 130° antegrade locking option can be considered.
11 Insertion of end cap topenlarge
An end cap may be used, depending on the final assessment of the nail length in relation to the tip of the greater trochanter.
The distance between the proximal end of the nail and the tip of the trochanter determines the appropriate length of the end cap, which is between zero and 20 mm, in 5 mm increments. The end cap is introduced using a hexagonal screwdriver.
The end cap is optional for the standard femoral nail. It is mandatory when a helical blade is used, since it locks the position of the blade.
Pitfall: bony overgrowth
The removal of the end cap may be very difficult if an exchange nailing is required. Bony overgrowth may require an extensive dissection of the greater trochanteric region. On the other hand, the end cap will have prevented tissue ingrowth and so, after its removal, the insertion of the extractor device is easier.
12 Wound closure and assessment of alignment topenlarge
The procedure ends with the closure of the fascia and the skin as separate layers.
Assessment of alignment
Before the patient is moved from the fracture table, rotation of the leg is observed clinically and compared to the contralateral leg.
With the femur now stable, it is possible to perform a thorough examination of the knee joint to rule out additional ligamentous injuries.