1 Principles topenlarge
Compression plating provides fixation with absolute stability for two-part fracture patterns, where the bone fragments can be compressed. Compression plating alone is typically used for simple fracture patterns with low obliquity, where there is insufficient room for a lag screw. Where the obliquity will permit, the addition of a lag screw, across the fracture and through the plate, enhances stability.
Compression plating can only be applied in an open procedure.
The objective of compression plating is to produce absolute stability, abolishing all interfragmentary motion.
Dynamic compression principle
Compression of the fracture is usually produced by eccentric screw placement at one or more of the dynamic compression plate holes.
The screw head slides down the inclined plate hole as it is tightened, the head forcing the plate to move along the bone, thereby compressing the fracture.
Plate position on the femur / tension band principle
As a general rule the plate should be positioned on the lateral aspect of the femur.
A plate acts as a dynamic tension band when applied to the tension side of the bone and when a cortical contact is present on the opposite side to the plate.
With vertical load, the curved femur creates a tensile force laterally and a compressive force medially.
A plate positioned on the side of the tensile force resists it at the fracture site, provided there is stable cortical contact opposite to the plate.
It is important to restore axial alignment, length, and rotation. That means, in a simple A type fracture, a direct reduction of the main fragments is required.
Reduction can be performed with direct reduction tools.
2 Reduction topenlarge
Subtrochanteric fractures present a particular problem in terms of fracture reduction and alignment. Due to the strong iliopsoas muscle pull, the proximal fragment is flexed and externally rotated and therefore difficult to control.
In an open plating technique a preliminary reduction can be undertaken to facilitate the final reduction.
Often joysticks are useful to derotate the proximal fragment.
3 Plate fixation to proximal fragment topenlarge
Guide wire insertion and verification of trajectories
Prior to insertion of the guide wires, the plate should be slightly overbent at the level of transverse fracture, so that, when the load screw in the distal fragment is tightened, the medial (trans) cortex is compressed first. This avoids gapping of the medial cortex, which would otherwise occur and result in cyclical motion, leading to excessive strain in the healing tissues, delayed union and fatigue failure of the plate.
The proximal femoral plate is anatomically shaped to match the profile of the upper femur. First, the plate is adjusted optimally to fit the proximal fragment. Through the two attached wire guides, the proximal 2.5 mm guide wires are inserted into the proximal fragment.
The positions of the guide wires are verified under image intensification in both planes (AP and lateral).
Screw length measurement
The correct screw lengths are determined by measuring the remaining guide wire length, using the dedicated measuring device.
Proximal 7.3 mm screw insertion
Cannulated 7.3 mm screws (locking or nonlocking) are inserted over the guide wires into the proximal fragment.
4 Plate fixation to distal fragment topenlarge
Insertion of first screw into distal fragment
If the overall reduction is found to be satisfactory, the first cortical screw is placed in the distal fragment near the fracture. The screw is placed eccentrically.
Alternative: articulated tension device
In cases of nonunion, the articulated tension device is helpful in producing compression.
To improve the compression further, the first screw in the distal fragment can be placed eccentrically.
The tension device is dismantled after two additional neutral screws are placed in the distal fragment.
Placement of additional screws in the distal fragment
In accordance with the preoperative planning, additional screws are placed in the proximal and distal fragments.
Pearl: osteoporotic bone
In osteoporotic bone the use of bicortical locking screws is advantageous.