1 Bent vs straight nails topenlarge
Interference with the rotator cuff footprint
The entry point for a bent nail may go through the bony attachment of the rotator cuff. This bony defect in the footprint cannot be reconstructed later.
The entry point for a straight nail lies under the rotator cuff so the nail is inserted through the rotator cuff. This requires an incision, which should be made in the line of the tendon fibers which can then be closed effectively by a side to side suture.
Fixation in osteoporotic bone, fifth anchoring point
Due to the bone density in osteopenic bone straight nails provide a better fixation in the proximal humerus in the region of their entry point. Bent nails run through the greater tuberosity which has a lower bone density compared to the superior humeral head.
Interference with fracture lines
In proximal humeral fractures which consist of a fracture of the greater tuberosity the trajectory of bent nails often passes through the fracture line between the greater tuberosity and the humeral head whereas straight nails penetrate the humeral head medial to the fracture line.
Fixation pattern of screws
Straight nails run more medial in the axis of the medullary cavity. Therefore, it is possible to perform a direct fixation of the lesser tuberosity through the nail.
2 General considerations top
Sequence of reduction
- Reduce and fix the tuberosity to the humeral head (thereby converting the 3-part fracture into a 2-part situation).
- Reduce the proximal humeral segment to the shaft with the nail and fix it.
Protect the greater tuberosity fixation while doing the second part.
Suture reduction and fixation of the greater tuberosity
Sutures in the rotator cuff tendon insertions aid manipulation, reduction, and temporary fixation of a proximal humerus fracture.
Traction on the sutures helps achieve reduction. When tied, they bring the fragments together and stabilize them.
Correct nail entry point
A precise entry point of the humeral nail is crucial. For straight nails, the correct entry point is in line with the axis of the humeral shaft. An incorrect entry site results in malreduction of the metaphyseal fracture.
It might be difficult or even not possible to access the correct entry point if the humeral head is displaced severely into a varus position. Therefore, it is strongly recommended to expose the entry point by manipulating the humeral head. K-wire “joy-sticks” (as illustrated) or sutures through the rotator cuff insertions can be used to achieve this.
Reduction of the metaphyseal fracture component
If the entry point has been chosen correctly, insertion of the nail will help reduce the fracture.
Protection of axillary nerve
The main structure at risk is the axillary nerve. The axillary nerve should be protected by limiting the incision to less than 5 cm distal to the acromial edge, by palpating the nerve, and by avoiding maneuvers that stretch the nerve during reduction and fixation.
In addition, any suspicious screw trajectory should be made to the bone with blunt dissection and checked with finger palpation if necessary.
Remember the course of the nerve when placing the screws depending on the chosen nail and/or position of the nail itself.
Tension band sutures in addition to nail
Sutures placed through the rotator cuff tendons and the bolts increase stability, and should be used as well as the nail and screws, particularly for more comminuted and/or osteoporotic fractures. With osteoporotic bone, the tendon insertion is often stronger than the bone itself, so that sutures placed through the insertional fibers of the tendon may hold better than screws or sutures placed through bone.
These additional sutures are typically the last step of fixation.
An anterolateral incision is recommended for nailing of proximal humeral fractures. The need for additional access depends upon fracture type. Extension (up to 6 cm) of the anterolateral incision may be needed.
AO teaching video: Proximal humerus – 3-part fracture – Reconstruction using the MultiLoc Proximal Humeral Nail
3 Reduction and preliminary fixation of the tuberosities topenlarge
Goals of reduction
The goal of reduction is to restore the anatomic position of all fracture components. For simple 3-part fractures, the humeral head and displaced tuberosity must be reduced properly.
The optimal reduction and fixation procedure for the fracture subtypes depends on the involved tuberosity, and whether or not the calcar region is comminuted.
In the following procedure, involvement of the greater tuberosity is assumed. For specific details on reduction of the lesser tuberosity see Open reduction and preliminary fixation of lesser tuberosity and humeral head.
In fractures with medial metaphyseal fragmentation, it is necessary to realign the medial column fragments as accurately as possible while preserving their soft-tissue attachments and vascularity. Once healed, the restored medial bony buttress helps prevent varus collapse. With medial fragmentation, initial fixation must be secure enough to resist varus collapse.
It is also crucial to reduce the humeral head adequately since the combination of remaining varus displacement and medial fragmentation predisposes to secondary varus collapse and/or implant failure.
Place rotator cuff sutures
Subscapularis and supraspinatus tendon
Begin by inserting sutures into the subscapularis tendon (1) and the supraspinatus tendon (2). Place these sutures just close to the tendon’s bony insertions. These provide anchors for reduction, and temporary fixation of the greater and lesser tuberosities.
Next, place a suture into the infraspinatus tendon insertion (3). This can be demanding, and may be easier with traction on the previously placed sutures, with properly placed retractors, and/or repositioning the arm.
Use of stay sutures
Anterior traction on the supraspinatus tendon helps expose the greater tuberosity and infraspinatus tendon.
Insert a preliminary traction suture into the visible part of the posterior rotator cuff ...
... and pull it anteriorly. This will expose the proper location for a suture in the infraspinatus tendon insertion. Then the initial traction suture is removed.
Pearl - Larger needles: A stout sharp needle facilitates placing a suture through the tendon insertion.
Reduction of the greater tuberosity
Direct reduction of the greater tuberosity fragment is performed by pulling the sutures or …
… with instruments (eg elevator) applied either through the incision (as illustrated) or through a separate stab incision.
Preliminary suture fixation of the greater tuberosity
Tighten and tie the transverse sutures in order to preliminarily fix the greater tuberosity fragment. Thereby, the 3-part fracture is converted into a 2-part situation.
Preliminary reduction of humeral head
Use of a K-wire
A “joy-stick” technique, as illustrated, may also be helpful to manipulate the humeral head.
4 Determination of entry point and opening of the canal topenlarge
Determination and exposure of entry point
The nail insertion site lies on the axis of the humeral shaft. It is located intraarticularly posterior to the tendon of the long head of the biceps within the cartilage. It is typically near the highest point of the humeral head. It is slightly anterior to the center of the greater tuberosity. A supraspinatus split is necessary.
Incise the supraspinatus tendon for approximately 3 cm over the foreseen entrance point posterior to the long head of the biceps. Retract the edges of the supraspinatus tendon with the help of stay sutures.
Insert a K-wire through the correct entry point and confirm proper placement by image intensification.
Open the humerus
A cannulated drill bit is recommended for opening the proximal humerus. This drill bit should be inserted over the previously placed guide wire. It should be advanced into the proximal medullary canal.
Alternatively, an awl may be used to open the humerus.
Take care not to displace the tuberosity fracture during passage of awl or drill.
5 Nail insertion topenlarge
Mount nail on insertion handle
The humeral nail is mounted on an insertion handle. The nail must be rotated correctly relative to the insertion handle.
Insert the nail with slightly rotating movements down to the metaphyseal fracture line. Pass the fracture zone under image intensification and make sure that the nail enters the distal fragment properly.
Make sure the proximal end of the nail is placed beneath the bony surface of the humeral head.
No protrusion of the nail may be tolerated. Confirm with appropriately oriented C-arm images that the nail is below the bone.
Be careful not to redisplace tuberosity fractures when inserting proximal locking components.
A K-wire placed through an aiming device may help to adjust the appropriate insertion depth.
Retrotorsion of the nail
In order to lock the nail in the correct trajectory, mount the aiming arm and swivel it approximately 25° anteriorly in order to follow the retroverted axis of the humeral head. (Due to the physiological retrotorsion of the humeral head, the axis of the humeral head is directed approximately 25° posteriorly to the condylar plane of the distal humerus.)
To obtain a true AP view of the proximal humerus, the forearm has to be rotated approximately 25° externally relative to the sagittal plane.
6 Proximal nail fixation topenlarge
The trocar for the planned screw fixation is placed through the aiming device, if necessary through a separate skin incision. Perform blunt dissection down to the bone so that the trocar may be placed without interposition of soft tissue.
Check once more the correct retrotorsion.
Drill with the calibrated drill bit carefully just to the level of the subchondral bone. The final position of the drill bit can be checked with the image intensifier. The screw length can be determined from the measurements of the drill bit.
Widen the screw hole in the lateral cortex
If necessary, a conical drill bit may be used to countersink the lateral cortex.
Attach the screw to the screw driver and insert the screw.
In some systems, as the one shown in this illustration, a screw-in-screw fixation is possible (see below). If so and if also intended, the plane of the handle of the screw driver should be horizontal and the arrow should point posteriorly once the final position of the screw is reached.
Pearl: Using an additional trocar, K-wires can help to achieve more stability temporarily.
Insert further screws
Insert further screws using the aiming device according to the fracture pattern.
Fix the lesser tuberosity
If necessary, fix the lesser tuberosity with an additional, anterior screw. Doing so, mount an anterior aiming arm to the aiming device and insert the screw as described above.
If necessary, use the screw-in-screw fixation for further stability.
Attach the centering sleeve to the already inserted screw, turn it anteriorly until it locks.
Insert the drill sleeve and predrill to the subchondral bone.
After length measurement insert the screw with the screw driver and torque limiter.
Calcar screw insertion
If necessary (such as varus fractures), insert a calcar screw through the aiming device in the way as described for the other screws.
7 Distal nail fixation topenlarge
Drill and determine length of locking screw
For distal locking, insert the trocar with the drill sleeve through one of the distal holes in the aiming device. Drill through both humeral cortices until the bit just breaks through the medial cortex and read the depth from the drill bit. Alternatively, a depth gauge can be used.
Insert a locking screw through the trocar. A second screw is recommended, especially in osteoporotic bone.
8 Finalizing the nailing topenlarge
Insert end cap
The end cap prevents tissue from plugging the inner thread of the nail.
End caps are available in different sizes and can, if necessary, be used to extend the nail. The top of the end cap must not protrude above the surface of the bone.
Repair rotator cuff
Suture the supraspinatus split.
Supplementary rotator cuff tendon sutures
Rotator cuff tension band sutures are placed through the dedicated holes of the bolts.